Sorry, there was a problem. Note that the written signal is in the form of an array. Guaranteed delivery available Payments: Customers who viewed this item also viewed. You can create a task for analog output in the same way, that is, by selecting Create Task in the right-click menu of the USB device as it appears under Devices and Interfaces , see Figure 8.
|Date Added:||2 November 2016|
|File Size:||37.32 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
TechTeach: National Instruments USB
Ni usb-6008 more Seller information optoscience Selecting Create Task in the menu opens a window where you can ni usb-6008 the measurement type, see Figure The configuration can be saved using the Save button, see Figure There’s a problem loading this menu right now.
Via the Polymorphic VI Ni usb-6008 you can select the type of data the function will write. Interest will be charged to your account from the purchase date if the balance is not paid in full usb-60008 6 months. There was a problem filtering reviews right now.
A list of options for the measurement type. Ni usb-6008 to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to watch list.
This opens the window shown in Figure ysb-6008 where you can select the measurement type from a list. Delivery times may vary, especially during peak periods.
Assume as an example that your application requires two analog inputs. This item will be shipped through the Global Ni usb-6008 Program and includes international tracking. In the folllowing it described how to create tasks for the analog input and the ni usb-6008 output operations. By clicking the Finish button in the window shown usg-6008 Figure 15, the window shown in Figure ni usb-6008 is opened where uxb-6008 can select the physical channels for your task.
Referenced single ended which means that all channels have the same ground. The test procedure, which is denoted loopbackis to connect the AI0 channel to the AO0 channel.
Above we created a ni usb-6008 for an analog input. Figure 5 shows the window thereby ni usb-6008 opened. Please enter a valid ZIP Code. Output low is below 0. One person found this helpful. Learn More – opens in a new window or tab.
If a global channel is reconfigured, all tasks which involve that global channel are updated accordingly. In this example, no feedback control system is implemented. ni usb-6008
A quick guide to National Instruments USB-6008
Please enter a number less than or equal to 2. Image Unavailable Image not available for Color: ComiXology Thousands of Digital Ni usb-6008.
Voltage range is 0 – 5V fixed. Write a customer review. The Create New Task dialog window You must define one task for each of the input or output operations, i. Back to home ni usb-6008.
United States and many other countries See details. Assuming that MAX recognizes the device, you can now run a self-test of it by selecting Self-Test in the menu. If you had written a value outside this range to the DAQmx Write function you would ni usb-6008 an error due to signal limitations function it can only generate voltage outputs ni usb-6008 0 and 5 volts. There may be a small difference between the ni usb-6008 due to the limited resolution in the DA-converter digital-to-analog and in the AD-converter analog-to-digital.
A task definition may include the following here an analog input task is assumed: